Approximately 18 x 23 x 23 cm (7 x 9 x 9 in), excluding pole clamp protrusion and power cord storage
Approximately 6.0 Kg (13 lb), with battery
110-120 VAC, 50/60 Hz, 30 W
Hospital-grade AC cord. 8 ft long, with transparent Plug and retainer plate
0.5 A, 250 V, Slow Blowing
One sealed, lead-acid, rechargeable 8 V battery, Internal to device. Accessible for ease of field replacement, with color-coded leads and polarized connector. Battery life (new batteries, full charge, at 20 C) approximately 500 mL total volume delivered or six hours of operation, whichever occurs first.
Delivery Rate Range
0.1 to 99.9 mL/hr (in o.1 mL. Increments) The total of the primary rate plus the secondary rate cannot exceed 99.9 mL/hr in Micro Mode. Note: while using the concurrent delivery feature in the Micro Mode, neither the primary rate nor the secondary rate can be less than 0.5 mL/hr.
1 to 999 mL/hr (in 1 mL increments the total of the primary rate plus the secondary rate cannot exceed 700 mL/) in 1 mL/hr in Macro Mode
Dose Limit Range
0.1 to 999 mL (in 0.1 mL increments)
1 to 9999 mL (in 1 mL increments)
-2 to 10 psig. The maximum pressure is user-selectable from 1 to 10 psig, via the front panel touch switches.
For dual-channel delivery, a single dose of a secondary fluid may be administered
For multidose delivery, 1 to 24 doses of a secondary fluid may be administered at intervals from 15 minutes to 24 hours
The Distal Occlusion alarm sounds within two pumping cycles after the distal set tubing or set outlet fitting becomes occluded
The Proximal Occlusion alarm sounds within five pumping cycles if the tubing proximal to the cassette becomes occluded
The Stopped Air In Distal Line alarm will sound if a bubble 100 uL or larger passes the distal air-in-line sensors (alarm may actually sound at detection of a bubble as small as 50 uL)
The Stopped Air In Proximal Line alarm will sound if approximately 1200 uL or more passes through the proximal air-in-line sensors